Continuous ambulatory pulmonary artery pressures were recorded using a conventional No 5 French Goodale-Lubin filled catheter linked to the Oxford Medilog system of a portable transducer-perfusion unit and miniaturised recorder. Data retrieval and analysis were performed using a PB2 Medilog playback unit linked to a PDP 11 computer system. The total system has a frequency response linear to 8 Hz allowing accurate pressure recording over the full range of heart rates. Ten recordings in 10 patients yielded artefact free data for 80% or more of the recorded period. This inexpensive reliable method allows pulmonary artery pressures to be recorded in unrestricted subjects.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.