An aneurysm of the fossa ovalis is frequently associated with an atrial septal defect. Intervention with transcatheter closure of such defects may be difficult since it bears the risk of inadvertent deployment of the device with both umbrellas in the right or left atrium. A novel technique for the closure of such multiperforated aneurysms of the fossa ovalis, which allows central positioning and controlled deployment of the device, is reported.
- fossa ovalis
- atrial septal defect
- transcatheter closure
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
To facilitate central crossing of the multiperforated aneurysm of the fossa ovalis, a balloon tipped catheter (Berman Wedge) was filled with diluted contrast medium within the right atrium and advanced under echocardiographic guidance into the aneurysm of the fossa ovalis, thus centring the end hole of the catheter within the aneurysm (fig 1A). A guidewire was advanced through the catheter into the left atrium; the balloon was deflated and also advanced into the left atrium and further into the left upper pulmonary vein. Transoesophageal echocardiography confirmed that the catheter crossed the centre of the multiperforated aneurysm of the fossa ovalis. During sizing of the defect, echocardiography demonstrated that the redundant tissue of the aneurysm was pushed into the left upper pulmonary vein.
Inadvertent deployment of both umbrellas of the CardioSeal device in the left atrium was avoided by using a double sheath technique. A 23 mm CardioSeal device was front-loaded into a 9 French long sheath which was pre-shaped like an NIH catheter using hot vapour. An 11 French long sheath was shortened, so that it was 6 cm shorter than the 9 French introducer sheath, and pre-shaped using hot vapour. The 11 French long sheath was advanced into the left upper pulmonary vein over the 0.035 inch guidewire. The dilator and guidewire were retrieved and the 9 French long sheath, which had been preloaded with the 23 mm CardioSeal device, was advanced to the tip of the 11 French sheath. The left umbrella of the device was opened within the left upper pulmonary vein, to assure its position on the left atrial side of the floppy aneurysm of the fossa ovalis (fig 1B). Thereafter, both sheaths were pulled back under echocardiographic guidance until the left atrial arms of the CardioSeal device were in contact with the interatrial septum. The 11 French long sheath was pulled back under echocardiographic guidance over the 9 French long sheath into the inferior vena cava, to assure that the tip of the 11 French long sheath was on the right side of the interatrial septum. The 11 French long sheath was then re-advanced towards the CardioSeal device to push the aneurysmatic tissue to the centre of the device (fig 1C). Approximation of the aneurysm tissue to the (opened) left atrial umbrella was confirmed by hand injections with contrast through the 11 French sheath and by echocardiography. The right umbrella of the CardioSeal device was opened into the 11 French sheath by pulling back the 9 French sheath. Echocardiography demonstrated that the aneurysmatic tissue was still kept in position by the 11 French long sheath. To deploy the right atrial arms the 11 French long sheath was pulled back. The right atrial umbrella was opened on the right side of the aneurysmatic fossa ovalis, resulting in complete closure of the multiperforated aneurysm of the fossa ovalis. The aneurysmatic tissue was caught within the device (fig1D). Echocardiography six months after the procedure showed complete occlusion of the multiperforated aneurysm of the fossa ovalis and capture of the aneurysmatic tissue between both umbrellas of the device, with no residual shunting.