Statistics from Altmetric.com
Indications for extraction of chronically implanted pacemaker leads have been classified as mandatory, necessary, or discretionary.1 ,2In published reports as well as in clinical practice, most indications cited are non-vital. These indications are often based on clinical judgement, even in published policy statements.3 We therefore reviewed published work on this subject, starting from the Pubmed database, in an attempt to provide an evidence base for the benefits of lead extraction. We begin with a brief overview of the results and complications of current extraction techniques.
Recently, comprehensive endovascular techniques have been developed for pacemaker lead extraction.2 ,4 ,5 In the superior vena cava approach, a locking stylet is introduced into the lead and locked close to the distal electrode in order to apply traction directly to the tip.6 If gentle traction is not successful, telescoping sheaths can be advanced over the lead to disrupt fibrous binding of the lead to veins or myocardium. When necessary, the tip of the lead is freed by countertraction, the sheath being positioned against the myocardium to prevent inversion during traction on the lead. Recently, a laser sheath has been introduced which uses photoablation to disrupt the fibrous bindings instead of mechanical force.7-10
In the transfemoral approach, the pacing lead is grabbed with a deflecting guide wire or retriever through a long sheath inserted from the femoral vein.7 ,11 The proximal end of the lead is pulled down from the subclavian vein. Then the outer sheath is advanced over the lead to disrupt the scar tissue, as with the superior approach. When the myocardium is reached countertraction is applied.
Table 1 summarises the results of the different extraction techniques. With the use of only locking stylets, Alt et al achieved total removal of 81% of 150 leads, partial removal of 12%, and failure in 7%.6 There were no major complications. In the US lead extraction database, extraction was attempted with either a conventional superior or an inferior approach.11 Complete removal was achieved in 86.8% and partial removal in 7.5% of 2195 leads. Major complications occurred in 2.5% of the 1299 patients. Eight patients died (0.6%). In a paper describing experience with lead extraction between 1994 and 1996, Byrdet al reported complete removal in 93%, and partial removal in 5% of 3540 leads. Major complications occurred in 1.4%, including one fatality.12
In a randomised trial, a laser sheath achieved 94% complete removal of the lead against 64% with conventional sheaths during initial attempts at extraction of 465 chronically implanted leads.8 In a registry of laser sheath extraction in 1463 patients, complete success was achieved in 90% of 2249 patients and partial success in 3%.10 There were complications in 3.4% of patients, including tamponade (1.4%) and haemothorax (0.3%). Twelve patients died (0.8%), mostly as the result of tamponade.
Byrd et al reported an increased risk of failed or partial extraction with increasing implant duration (doubling every three years).12 Implant duration was not linked to complications in any of the cited papers. However, major complications of extraction of 2732 Accufix leads (Telectronics Pacing Systems Inc, Englewood, Colorado, USA), reported by the Accufix Research Institute on www.accufix.com, increased from 2% at one year implant duration to 8.3% with an implant duration of more than five years. Physician's lack of experience, increasing numbers of leads extracted, and female sex have, however, been associated with an increased risk of complications.11 ,12
There are no randomised studies comparing lead extraction with a conservative strategy aimed at salvaging the leads and the generator. Some reports have compared both treatment options in a large patient population from a single centre.
Lewis et al described 75 patients (including 10 with epicardial systems) presenting with pocket abscess or erosion, including 17 patients with positive blood cultures.13 From 32 patients treated conservatively (antibiotics, limited debridement, and irrigation or aspiration of the infected site) only one patient was cured. In the other 43 patients, primary removal of all hardware led to successful resolution of infection in all. Also, the 31 patients who failed conservative treatment were cured after removal of the pacing system. Molina reported the outcome of infection in 21 pacemaker and 17 defibrillator patients.14 Infection persisted in all 12 patients treated with antibiotics only (including two deaths). Primary removal of leads and generator in 26 patients cured all of them. These data are supported by reports of primary removal of the pacing system. Brodmanet al had a successful outcome in 41 of 42 patients in whom all hardware was removed.15 One patient died postoperatively because of persistent sepsis from a retained lead segment. Klug et al tried to remove all material either endocardially or surgically in 52 patients with pacemaker related endocarditis.16 Two patients died after surgical treatment. Infection reoccurred in two patients, and three patients died late after the procedure with symptoms of an infective syndrome.
Although successful conservative treatment of infection is described in many case reports, only a few papers mention success in a substantial number of patients. Hurst et al described successful eradication of infection in 19 patients with infection limited to the pacemaker pocket or skin erosion.17 Treatment consisted of extensive debridement and generous enlargement of the pocket to ensure adequate closure without tension. The pocket was irrigated with a closed irrigation system containing antibiotics and tyloxapol. Garcia-Rinaldiet al successfully relocated the generator to a deeper subfascial plane in the abdominal wall in 10 patients with exposed pacemakers without gross signs of infection.18
It can be concluded that total removal of the pacemaker system is the most reliable way of eradicating any pacemaker related infection, and most clinicians agree on extraction in the event of systemic infection. Attempts at conservative treatment should be limited to patients presenting with skin erosions or low grade pocket infection. If chosen, relocation of the pacemaker to a deeper plane is advisable, along with careful attention to alleviating skin erosion by any hardware. If the attempt at conservative treatment fails, complete extraction is advised.
Whenever extraction is attempted, complete removal of the pacing system is important. Persistence of infection has been described in 17–77% of patients in whom only the generator was removed.19-22With this approach, Parry et al reported three deaths among 53 patients and Rettig et al four deaths among 21 patients.
Failure to extract the entire lead can lead to persisting infection. Lewis et al needed to do a thoracotomy to remove a retained lead segment in three of four patients with failed endovascular extraction before infection was controlled.13Molina had to reoperate on two patients with a retained lead before infection was controlled.14 There is also concern in pacemaker related endocarditis over embolisation of vegetations adhering to the lead when endovascular extraction is attempted. Kluget al removed leads intravascularly in 38 patients with vegetations of less than 10 mm.16 Although pulmonary scintigraphy in 33 patients showed a 30% new embolisation rate, these were without clinical sequelae. Surgical extraction was performed in 10 patients with larger vegetations. In this group, two deaths occurred after septic complications. This is an argument for expanding endovascular lead removal to include patients with larger vegetations, as there is no evidence of a deleterious effect of endovascular extraction compared with surgical extraction.
It is useful to reconsider the indication for pacing after successful extraction of the infected pacing system. Discontinuation of pacemaker treatment after lead extraction is reported in 13–52% of patients.13 ,23 ,24 In these patients, there is no advantage in salvaging a non-used pacing system.
Risk of venous occlusion
Meyers et al and Byrdet al stated that if too many leads are present, the resultant reaction has a high probability of obliterating the vein.1 ,2 As it is, non-functional leads make up 31–40% of indications for lead extraction.6 ,11 ,12
However, abandoned non-infected leads, with the wires insulated and the lead securely fastened to the fascia, have a very low complication rate. Furman et al and Parryet al reported complications in, respectively, one of 152 and two of 119 patients with abandoned leads.21 ,25 In a recent prospective study, de Cocket al found no differences with respect to thrombotic complications in 48 patients with multiple leads (39 patients with three leads, nine with four leads) versus 48 age matched control patients with dual chamber pacemakers.26 Of all phlebographic studies, only Pauletti et aldescribed an increased incidence of occlusion with multiple (mostly two) leads compared with single leads (26%v 8%).27-33
In contrast to an incidence of between 8% and 21% of occlusion of the subclavian or brachiocephalic vein in these studies, the incidence of pacemaker induced superior vena cava syndrome is reported to be very low, at 0.03–0.4%.34-36 In a review by Mazetti et al reporting 37 cases of superior vena caval obstruction, only five patients had one or more abandoned but intact leads.36 Thus it is unlikely that properly abandoned leads are a prime risk factor for occlusion.
It is important to realise that in contrast to an intact abandoned lead, infected or severed leads after failed extraction attempts do constitute a risk factor for venous occlusion. Infection was present in 30% and a severed lead in 43% of the 37 patients described by Mazetti et al, including nine with both infection and a severed lead.36 Zerbeet al reported superior vena cava syndrome in two patients after previous extraction attempts.37Mitrovic et al performed phlebography in 12 patients with right sided severed leads in whom the free distal end was left unsecured intraluminally.30 Complete superior vena caval obstruction was observed in two patients, all others having an occluded brachiocephalic or subclavian vein. In contrast, only 15 of 100 consecutive patients with normal functioning leads had venous occlusion.
In conclusion, there is no evidence that properly abandoned leads are a risk factor for venous occlusion and they should not be routinely extracted. Although an argument is sometimes made that an “excessive” number of leads will inevitably give rise to complications, there is no published evidence to support this. Further investigation into the natural history of patients with multiple leads, and also into the thrombogenicity of the extraction procedure itself, is necessary before lead extraction can be considered beneficial with any number of leads present.
Although there are no reports specifically addressing non-functional leads in young patients, special considerations may apply to this population—for example, the likely need for multiple implants in the future, long implant duration, and small body size. From the viewpoint of lead extraction, timely extraction may increase the success rate and avoid the complications of having to extract multiple leads with a longer implant duration in the future. In particular, the lack of proof that abandoned leads are harmful does not justify indiscriminate extraction of non-functional leads, even in young patients. Furthermore, a smaller body size may increase the risks of lead extraction—the higher complication rate described in female patients could well be related to body size, so the same may also be true for young patients.11 ,12 A sensible approach therefore may be to extract only leads with a short implant duration using technology adapted to the body size.
An exception to this recommendation may be pacemaker related venous obstruction, where lead extraction can result in recanalising the occlusion. Ing et al first extracted the malfunctioning lead that caused superior vena caval or innominate vein obstruction in two patients.38 They then passed a guide wire through the obstruction and dilated and stented the vein, after which they implanted a new lead. We used a laser sheath in three patients with venous occlusion to extract dysfunctional leads and cross the obstruction.39 We then inserted a guide wire beyond the obstruction to secure access to the central venous circulation before removing the sheaths. Subsequently, new leads were implanted by the same route. In this way, the risk of bilateral occlusion is avoided and the contralateral site is saved as an entry point in case of future complications.
Although Seeger and Scherer showed new pulmonary emboli on pulmonary scintigraphy after pacemaker implantation in three of 20 patients not treated with anticoagulation,40 clinically evident pulmonary embolism is rare—only one of 520 patients in the report by Porath et al.41 Furthermore, other predisposing factors are often present.42Anticoagulation is usually sufficient to prevent further embolisation.41-45 However, if embolisation persists despite anticoagulation, the mortality can reach 75% if the lead is not removed.41
Accufix and Encor leads
Protrusion of a fractured J shaped retention wire of Accufix and Encor bipolar atrial leads (Telectronics Pacing Systems) has been implicated as the cause of death in only six of 30 357 patients. The probability of remaining free of injury is 99.9% at five years after implantation, the rate of injury being linear or decreasing slightly as time from implant increases.46 However, lead extraction might have prevented injury in high risk patients: eight years after implantation, only 40% of the surviving population still had the Accufix lead in place.47 On the other hand, major complications of lead extraction—including 15 fatalities—occurred in 4.5% of 4076 patients.48 Also, as reported by the Accufix Research Institute on www.accufix.com, major complications of extraction increased from 2% at one year of implant duration to 8.3% with an implant duration of more than five years.
In conclusion, if the retention wire is intact, serial fluoroscopy to screen for the integrity of the J wire is appropriate. Extraction is advised if the J wire is protruding or has migrated, as this has the highest potential to cause injury. In patients with a suspected fracture and a longer life expectancy, extraction can also be advised. In elderly patients or those with serious comorbidity, serial screening can be an alternative in this instance. Where elderly patients present with a protruded or migrated J wire, decisions should be made on a case by case basis.
Endocardial leads inadvertently implanted in the left ventricular cavity should be removed immediately after implantation. If already chronically implanted, anticoagulation is reported to be efficient as primary or secondary prevention of cardioembolic stroke, although some authors prefer extraction.24 ,49-51 In our opinion, extraction should be reserved for recurrent embolisation occurring despite adequate anticoagulation, or where anticoagulation cannot be used. A surgical approach is preferred because of the risk of systemic embolisation with advancement of sheaths over the lead, even though Trohman et al described percutaneous extraction in a patient on chronic warfarin treatment without complications.52
Tricuspid valve insufficiency is sometimes cited as an indication for lead extraction.8 ,53 However, the outcome of extraction on valvar function is never mentioned.8Furthermore, we have found significant new regurgitation jets after ventricular lead extraction with a laser sheath in four of 14 consecutive procedures, as a result of direct injury to the tricuspid valve leaflets.54
Current intravascular extraction techniques are effective for extraction of pacemaker leads. However, there are still complications to be taken into account and there is a clear learning curve.7 ,12 Published reports provide solid support in favour of lead extraction in cases of infectious complications. Lead extraction as primary treatment, even for localised infections, will result in the best opportunity to cure the infection and prevent systemic complications. The indications for lead extraction are summarised in table 2.
In contrast, although there is a tendency to extract all malfunctioning superfluous leads during revision, the current complication rate of endovascular extraction techniques still outweighs the small risk of leaving the leads properly secured in the patient. The outcome of lead extraction of Accufix or Encor leads shows that, although various lead related complications might have been prevented, the risk of indiscriminate extraction can outweigh the inherent risk of the targeted complication. Therefore the possibility of extracting a lead may not necessarily be accompanied by an indication for lead extraction.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.