Statistics from Altmetric.com
- Myocardial perfusion scintigraphy
- coronary artery disease
- coronary physiology
- endothelial function
- heart failure
- myocardial ischaemia
- clinical coronary heart disease
- non-invasive assessment
- myocardial viability
- stress testing
- diagnostic imaging
- radionuclide imaging
- myocardial perfusion
- nuclear cardiology
Myocardial perfusion scintigraphy (MPS) with single photon emission CT (SPECT) has served the cardiology community for almost three decades. Significant developments have taken place over recent years covering almost every aspect of the technique. This article describes the technical innovations in scanners, software, imaging protocols and stress regimens and discusses its evolving clinical role.
Planar MPS was introduced in the 1970s. Two decades later, most centres have adopted SPECT with ECG gating for the combined assessment of myocardial perfusion and left ventricular function using thallium-201, technetium-99m-sestamibi or technetium-99m-tetrofosmin. Over the past few years MPS has evolved from a diagnostic test of high accuracy for the detection of coronary artery disease (CAD) (mean sensitivity and specificity 90% and 75%, respectively) to an important tool for risk stratification, for which a large body of evidence exists documenting its clinical value and cost-effectiveness in a broad range of patient subgroups.1–3 Recent innovations in the field are likely to enhance the value of MPS even further. Here we discuss the most important of these new developments.
Recent technological developments
A new generation of cameras has been developed that exhibits a remarkably high sensitivity compared with conventional gamma camera systems. These are steady-state ‘small-footprint’ cameras that use new cadmium zinc telluride (CZT) detector technology. In contrast to traditional sodium iodide (NaI) detectors, CZT detectors can translate photon energy and location directly into electronic pulses without the need for photomultiplication of scintillation events. The D-SPECT camera (Spectrum Dynamics, Caesarea, Israel) comprises nine rotating pixilated detector columns of …
Funding This work forms part of the research themes contributing to the translational research portfolio of Barts and the London Cardiovascular Biomedical Research Unit which is supported and funded by the National Institute of Health Research.
Competing interests None.
Provenance and peer review Commissioned; internally peer reviewed.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.